Nitric oxide (NO) is recognized as the most important small signaling molecule in the human body. An imbalance of NO is closely associated with many serious diseases such as neurological disorders, cardiovascular diseases, chronic inflammations and cancers. Herein two chemiluminescence-based devices (a real-time NO measurement device and a controllable NO delivery device) were developed to facilitate the NO quantitative study and obtain information for NO related drug design.
The first device used for real-time measuring NO(g) flux from living cells was developed and validated. The principle was to use a two-chamber design, with a cell culture chamber and a gaseous sampling chamber separated by a selective PDMS-based (polydimethylsiloxane-based) membrane. The membrane was polydopamine coated to improve the cell adhesion and growth. The signal response of the devices was validated by using the controlled NO releasing polymer SNAP-PDMS (S-Nitroso-N-acetyl-D-penicillamine covalently linked to PDMS). The real-time NO generation profiles of the macrophage cell-line RAW264.7 stimulated by lipopolysaccharide (LPS) and/or interferon-γ (IFN-γ) were successfully measured. Data also shows that the change of NO generation by different exogenous factors can also be tracked in real-time by the device. The maximum NO flux characterized by the device varies from approximately 2.5 to 9 pmol/106cell/s under 100 ng/ml LPS and 10 ng/ml IFN-γ stimulation depending on different culture conditions, indicating the current NO reporting method by using one single NO flux value is not sufficient to represent the dynamic character of NO in biological samples. By using the similar principle and device characterization processes, an innovative cross-membrane delivery system was developed to successfully deliver NO to the cultured cells in a dose and temporally controlled manner. Different levels and durations of NO were delivered to the smooth muscle cell line MOVAS cultured in the device to validate the controllability of the NO delivery. The device characterized an NO flux of 1.0x10–10mol/cm2/min as a threshold level, above which NO may cause MOVAS death, and the effect of this lethal level was also duration and cell-type dependent. To further show the utilization of the two devices, the NO measurement device was used to systematically study different NO donors' potencies. It is suggested that measuring the real-time NO profiles of specific NO donors in specific biological conditions can help us avoid generating confusing data and understand the potencies of NO releasing drugs.
Current methods for NO quantitative experiment lack the means to understanding the time-related aspects of NO in biological environment and controlling the actual NO level and duration that cells experience. The two innovative devices solved these problems in cell culture models, achieving quantitatively studying the behavior of NO in the biological conditions and potentially assisting the design of the NO releasing drugs/materials.
|Advisor:||Frost, Megan C.|
|Commitee:||Goldman, Jeremy, Lee, Bruce P., Tang, Xiaoqing|
|School:||Michigan Technological University|
|School Location:||United States -- Michigan|
|Source:||DAI-B 77/06(E), Dissertation Abstracts International|
|Subjects:||Biochemistry, Biomedical engineering|
|Keywords:||Controllbale, Delivery, Donor, Measurement, Nitric oxide, Real-time|
Copyright in each Dissertation and Thesis is retained by the author. All Rights Reserved
The supplemental file or files you are about to download were provided to ProQuest by the author as part of a
dissertation or thesis. The supplemental files are provided "AS IS" without warranty. ProQuest is not responsible for the
content, format or impact on the supplemental file(s) on our system. in some cases, the file type may be unknown or
may be a .exe file. We recommend caution as you open such files.
Copyright of the original materials contained in the supplemental file is retained by the author and your access to the
supplemental files is subject to the ProQuest Terms and Conditions of use.
Depending on the size of the file(s) you are downloading, the system may take some time to download them. Please be