Cavity ringdown spectroscopy (CRDS) makes use of light retention in an optical cavity to enhance the sensitivity to absorption or extinction of light from a sample inside the cavity. When light entering the cavity is stopped, the output is an exponential decay with a decay constant that can be used to determine the quantity of the analyte if the extinction or absorption coefficient is known. The precision of the CRDS is dependent on the rate at which the system it acquires and processes ringdowns, assuming randomly distributed errors. We have demonstrated a CRDS system with a ringdown acquisition rate of 1.5 kHz, extendable to a maximum of 3.5 kHz, using new techniques that significantly changed the way in which the ringdowns are both initiated and processed. On the initiation side, we combined a custom high-resolution laser controller with a linear optical feedback configuration and a novel optical technique for initiating a ringdown. Our optical injection "unlock" method switches the laser off-resonance, while allowing the laser to immediately return to resonance, after terminating the unlock, to allow for another ringdown (on the same cavity resonance mode). This part of the system had a demonstrated ringdown initiation rate of 3.5 kHz. To take advantage of this rate, we developed an optimized cost-effective FGPA-based data acquisition and processing system for CRDS, capable of determining decay constants at a maximum rate of 4.4 kHz, by modifying a commercial ADC-FPGA evaluation board and programming it to apply a discrete Fourier transform-based algorithm for determining decay constants. The entire system shows promise with a demonstrated ability to determine gas concentrations for H2O with a measured concentration accuracy of ±3.3%. The system achieved an absorption coefficient precision of 0.1% (95% confidence interval). It also exhibited a linear response for varying H2O concentrations, a 2.2% variation (1σ) for repeated measurements at the same H2O concentration, and a corresponding precision of 0.6% (standard error of the mean). The absorption coefficient limit of detection was determined to be 1.6 × 10-8 cm -1 (root mean square of the baseline residual). Proposed modifications to our prototype system offer the promise of more substantial gains in both precision and limit of detection. The system components developed here for faster ringdown acquisition and processing have broader applications for CRDS in atmospheric science and other fields that need fast response systems operating at high-precision.
|Commitee:||Atkinson, Dean, Khalil, Aslam, Lasseter Clare, Tami, Sanchez, Erik|
|School:||Portland State University|
|School Location:||United States -- Oregon|
|Source:||DAI-B 76/08(E), Dissertation Abstracts International|
|Subjects:||Physical chemistry, Optics|
|Keywords:||Cavity rindown, Laser spectroscopy, Optics|
Copyright in each Dissertation and Thesis is retained by the author. All Rights Reserved
The supplemental file or files you are about to download were provided to ProQuest by the author as part of a
dissertation or thesis. The supplemental files are provided "AS IS" without warranty. ProQuest is not responsible for the
content, format or impact on the supplemental file(s) on our system. in some cases, the file type may be unknown or
may be a .exe file. We recommend caution as you open such files.
Copyright of the original materials contained in the supplemental file is retained by the author and your access to the
supplemental files is subject to the ProQuest Terms and Conditions of use.
Depending on the size of the file(s) you are downloading, the system may take some time to download them. Please be