Dedicated breast PET scanners have been intensively investigated in recent years as their increased sensitivity and resolution are hypothesized to result in improvements in the early detection and determination of treatment response for breast cancer. However, there is currently no well-defined protocol for accurately evaluating their performance. This work aims to determine parameters for a set of phantoms for estimating count rate performances in pendant breast PET scanners.
A collection of 216 dedicated pendant breast CT images were divided into quartiles of overall volume (v), average radius ( r), and total length (l) of the breast. The 25th, 50th, and 75th quartile values of r and l were the parameters used for a set of 3 cylindrical phantoms. The cylindrical shape was chosen to remain consistent with the current NEMA whole body and small animal protocols. These cylinders, made out of high density polyethylene, are the small, medium, and large breast phantoms. For each phantom, the line source is placed at a radial offset of 4/5 of the phantom radius. Out-of-field of view activity is modeled using the NEMA-NU2 2007 scatter phantom. The parameters for each cylindrical phantom are as follows: Small, r = 4.2 cm, l = 7.1 cm; Medium, r = 4.8 cm, l = 9.0 cm; Large, r = 5.4 cm, l = 10.3 cm.
The activity ratios were determined by matching the patient prompts minus randoms (P − R) rate in each breast phantom and the remaining patient randoms (R) rate in the whole body phantom. The ratio of activity in the whole body phantom to activity in the breast phantoms is as follows: Small, 105:1; Medium, 40:1; Large, 22:1.
|Advisor:||Badawi, Ramsey D.|
|Commitee:||Boone, John M., Cherry, Simon R.|
|School:||University of California, Davis|
|School Location:||United States -- California|
|Source:||MAI 51/03M(E), Masters Abstracts International|
|Subjects:||Biomedical engineering, Medical imaging|
|Keywords:||Breast pet/ct, Dedicated breast pet, Pet breast phantoms, Pet imaging|
Copyright in each Dissertation and Thesis is retained by the author. All Rights Reserved
The supplemental file or files you are about to download were provided to ProQuest by the author as part of a
dissertation or thesis. The supplemental files are provided "AS IS" without warranty. ProQuest is not responsible for the
content, format or impact on the supplemental file(s) on our system. in some cases, the file type may be unknown or
may be a .exe file. We recommend caution as you open such files.
Copyright of the original materials contained in the supplemental file is retained by the author and your access to the
supplemental files is subject to the ProQuest Terms and Conditions of use.
Depending on the size of the file(s) you are downloading, the system may take some time to download them. Please be